BW April 2018

4/2: I sanded several thick fibers in bundle 405 to get them through the chimneys more easily. Andrew put the fibers back in the collar and aligned them before I left, but we still wanted to improve the alignment. I recorded the new configuration and tried to improve the alignment, but a Kapton strip broke and after I replaced it I did not get a very good alignment.

I also practiced fusing today.

4/4: This week I started polishing a fourth bundle of sci-fis. I am still twisting right in front and left in back. The wheel is running in reverse.

Yesterday during fusing practice the hook on one of the 500 g weights broke, and today Jim suggested wire as a replacement.

I worked on preparing light guide bundle 403 for testing today.

4/5: After several attempts by Andrew and me, I put bundle 403 in the dark box with a satisfactory alignment. I started running tests.

I changed the 400 grit sandpaper, and polishing went much faster. Bundle 4 is now less than 0.005" from the target length of 0.787". However, it is hard for me to tell where we bottom out - i.e., which is the smallest feeler gauge that prevents the bundle from touching the sandpaper. There seems to be a gradation of contact between the fibers and the grit, with a thick feeler gauge such as 0.025" allowing slight brushing contact while 0.015" allows the paper to scrape vigorously against the fibers.

4/6: Dr. Jones said it would be good for me to start fusing soon. We won't throw away fibers if fusing goes wrong. We'll just cut them and polish them again.

When the fibers are all tested, I need to come up with charts showing bell curves of all the test results. There should be one peak. However, I would expect two because of the difference between offset fibers.

I asked about telling which end of a sci-fi is highly polished, and Dr. Jones suggested using a microscope. I responded that it would probably be faster to polish both ends of the fibers. He said that would be all right if it doesn't take too long.

4/10: We may need some more fine-grit sandpaper. Ideally I want each sheet to suffice for four sci-fi bundle ends, which means two bundles if we polish both ends of the bundles and four bundles if we polish only one end of the bundles. However, it might sometimes take more than one circular piece to polish one bundle end, so it's more likely that I will need more sandpaper than less. I may need more 5 micron grit too to sand fibers for the chimneys.

4/11: I am not satisfied with the results of using a razor blade to remove wings from fused fibers. I keep making holes in the cladding. I want to compare using a razor blade with using 5 micron grit. I marked the fibers I used a razor blade on with purple masking tape, and I will leave the ones I sand unmarked.

I think I would still use the razor blade for big wings, but once they are cut short I would use the sandpaper. I have to experiment with sanding to be sure.

4/12: This week, after much practice, I started doing real fuses. They are meeting the strength requirement to hold 850 grams, but they might be too thick. I don't trust the thickness measurements because the wings are probably making them too large.

Since the fibers are normally strong enough, my main concerns about fusing are how thick the fibers are, how to remove wings without damaging the fibers, and how to avoid unwanted dust and material between the fibers to be fused.

I have sometimes removed dust by wiping the end of a fiber with my glove. That seems to work well. Wiping it with a lint-free cloth also works but can leave a thread behind.

A Lesson On Fusing
Jim said I should score the wing before removing it. That will help to guide the cut so the razor blade does not nick the cladding. He also suggested using sandpaper, maybe even 400 grit, on some of the fibers, but Dr. Jones said I should not do any sanding to remove wings. I tried using 5 micron grit on some of the practice fibers, but the fine grit was little help for removing wings.

Jim looked at my fused fibers and saw that the wings were not only at opposite corners but also at adjacent corners. He realized that one of the quartz ferrules is probably cracked.

He said the cladding has a lower boiling point than the core of the fiber, so sometimes it can boil away in spots. There can be a bubble that pops and leaves a little pothole, for example.

I should remove the wings before weight-testing the fibers because they could be strong when the rest of the fuse is weak.

To remove the wings, I should put the fiber down and run the point of the razor along the side until it cuts through the wing. I could also try sandpaper and look with the microscope or camera at the results.

I need to do some test fuses to find out where the crack in the ferrule is. The fibers are marked with masking tape on one side. That side will be facing up and towards me as I fuse. I also marked one fiber with a Sharpie, but that did not go as well and the mark should be disregarded.

4/16: I was sanding a practice fiber this morning and with the microscope I noticed sanding marks extending beyond the wings on the "sci-fi" end as well as on the light guide end. Either my sanding this morning was going too far or I had made the marks on a previous occasion of sanding. I think it is this morning's sanding that is responsible for the marks.

4/18: This week I have been trying to photograph fibers as Jim requested. I have not yet found out how to get good photos. Andrew got better ones than I did.

Dr. Jones found out that there was not enough room on the computer drive that stores test data, so he moved some data off the drive today. He looked at the jumps in the data on the second test for bundle 404 and said that is why we take multiple tests. If one is off, we have others that give a better picture.

4/20: Today Andrew and I got bundle 406 ready for testing, and I ran some tests in the afternoon.

I finished polishing sci-fi bundle 5 and went back to bundle 1 since I mistakenly decided to redo the polishing last month. Now I want to polish both ends highly, and I made progress toward that goal. I polished both ends with 1 micron grit today and polished one end with 0.3 micron grit. The next step for polishing bundle 1 is to polish the other end with 0.3 micron grit. The bundle is in the right position on the sci-fi guide to do that.

4/24: Today I finished polishing sci-fi bundle 5 and took the fibers out of the collar. When I start fusing again, I should use sci-fis from the other bundle that is out of the collar rather than bundle 5 because I used some fibers from the other bundle already. I don't want to mix up bundles. I want to know which sci-fi and light guide bundles are fused together.

4/26: I need to dry the metal pieces, scrub them with a wire brush, wash them with soap and water, and put them on a paper towel under the fume hood. Later under the fume hood I can spray ethanol on them and in the screw holes. Then I can leave the pieces on another paper towel.

I also need to wash the beaker with detergent and water once the gasoline has evaporated from it.

4/27: Today I got as far as putting ethanol on the metal pieces. I may be done with them now. I still need to wash the beaker. I want to make sure the gasoline evaporates as much as possible from the beaker.